Unlike some systems that require you to scroll through multiple numbers to code the meter manually, the code chip for the Accu-Chek Aviva Nano blood glucose. View and Download Accu-Chek Aviva Nano user manual online. Aviva Nano Blood Glucose Meter pdf manual download. Page 1. Page 2. Page 3. Page 4. Page 5. Page 6. Page 7. Page 8. Page 9. Page Page Page Page Page Page Page Page Page
|Published (Last):||12 February 2010|
|PDF File Size:||2.87 Mb|
|ePub File Size:||16.44 Mb|
|Price:||Free* [*Free Regsitration Required]|
The meter will not turn on or the display is blank. Do not move the insulin pump or the cable during the communication between the insulin pump accu-chem the computer. Press the lancing device 6. Remove the control solution you want to surface, like a aviga. Accu-Chek Performa Connect Click for instructions. Blood obtained from a fingertip or palm can be used at any time to measure blood glucose.
Lance your fingertip with the lancing device. Page 15 Yellow Release Button 5. I need to order some self test diary but I can’t order online can you let me know how I can get some. Accu-Chek Performa Nano Click for instructions. Doing so puts the Accu-Chek Go meter in communication mode. Not all devices are available in every country. Page 85 All other product names and trademarks are the property of their respective owners.
The meter is in set-up The meter is ready for a mode, waiting for you drop of blood or control to change or confirm solution.
Understanding Out-of-range Control Test Results Understanding Out-of-Range Control Test Results If the control result is not inside the acceptable range, here are some things you can do to solve the problem: Accu-Chek Performa Combo Click for instructions. Do not use this meter close to sources of strong electromagnetic radiation. mankal
Devices Available Outside the United States
Accu-Chek Aviva Expert Click for instructions. If you agiva have problems, call Accu-Chek Customer Care at Before attempting to connect the Accu-Chek Aviva meter for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:.
You may need to change the time to your time zone. The control manuak is printed on the bottle label. Don’t have an account? Press and hold until 3. Blood samples flashing blood drop taken from the palm are equivalent to blood taken from the symbol appear on fingertip.
Devices Available Outside the United States
Blood glucose may A pre-meal marker was be higher than accu-chke assigned avcu-chek this test measuring range of result. Accu-Chek Instant Click for instructions. If you want to mark the test result as a pre-meal or post-meal test, or with a general marker, leave the test strip in the meter see the next section. To perform a blood glucose test from an alternate site, you need the meter, a test strip, and a lancing device with the clear AST cap. Please refer to the user’s manual that came with the devices for instructions on how to pair the meter and pump.
Accu-Chek Aviva Nano User Manual
Specifications Blood volume 0. Page 19 Here is how to mark a test result and initiate a post-meal reminder: The meter needs little or no maintenance with normal use.
The built-in infrared window in the bottom of the Accu-Chek D-TRONplus insulin pump allows for communication between the maanual pump and the computer. High blood glucose hyperglycemia: Follow the infection control procedures appropriate for your facility. Before attempting to connect the Accu-Chek Performa meter for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:.
Viewing Test Results 7, 14, 30, and 90 day averages. Before attempting to connect the Axcu-chek Spirit insulin pump for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:.
Coding the Meter 1. Page 67 A pre-meal marker Blood glucose is was assigned to below the defined this test result hypoglycemia and the post-meal low blood reminder has been glucose level.
The result appears on the display. Accu-Chek Compact Plus Click for instructions.
ACCU-CHEK AVIVA NANO USER MANUAL Pdf Download.
To avoid electrostatic discharge, do not use the meter in a very dry environment, especially one in which synthetic materials are present.
Install the software according to the instructions. Viewing Test Results With the meter on or off, press and release to enter memory.
Accu-Chek Instant S Click for instructions.
General Average Past Results Press